- The units circulate heated air over the skin to provide a closed, controlled environment that warms an infant.
- The temperature range of the incubator in the manual mode should be at least 2
- Alarms should sound in the following events: high air temperature, low air temperature, high skin temperature, low skin temperature, fan failure, sensor failure, and power failure. A backup thermostat should be available, and shielding if a light bulb
The Units that are designed for transport should provide thermal support for infants during transport within and between hospitals even during adverse weather conditions.
- When operating on internal batteries, units should maintain air temperature.
Copyright © 2019 | Americare Medical Distributors Ltd. All rights reserved